|Chamber Door||Manual Hinge Door|
|Chamber type||Cylindrical chamber|
|Gravity||WM-360:Manual gravity/WM-380:Automatic gravity|
|Cycle progresses|| WM-360:Manually by open/close manual valves through conditioning、sterilizing 、exhaust、vacuum phases.|
WM-380:Cycle progresses automatically by control system through conditioning、sterilizing、exhaust、vacuum drying and venting phases.
|Sterilize temperature|| WM-360:is adjustable from 121℃ to 132℃ on temperature regulator.|
WM-380:is adjustable from 120℃ to 132℃ on temperature regulator. When setting temperature is achieved the process of sterilizing cycle will automatic operating.
|Sterilize time and drying time|| WM-360:is adjustable from 0~60 minutes on timer.|
WM-380:is adjustable from 0~60 minutes on timer. The timer starts only after the pre-set temperature is reached, ensuring complete sterilization.
|Illuminated pilot lamps||display status and a buzzer announces the completion and cuts off the beating power source.|
|Chamber pressure gauge/Jacket pressure gauge/Thermometer||make ideal require reading and indicate high precision and consistent results.|
|Integral Steam Generator||equips with a high efficient immersion beater and temperature to provide strict control of heating temperature.|
|Vacuum drying device||to ensure efficient drying function.|
|Automatic air venting device and steam trap||discharges cool air and condensate from chamber to obtain effective saturated steam.|
|Air Filter||An effective HEPA air filter is used to prevent contaminated air from entering into the chamber.|
|Chamber door||is manually operated and actuated by rotating a door handle.|
|Chamber material||is made of high quality SUS316 stainless steel with polished surface for long lasting service and easy cleaning. A double layer construction is thermally insulated with fiberglass and cover with stainless steel sheet.|
|Door Gasket||The silicone-rubber sealing gasket is applied to an even pressure door seal.|
|Pressure vessel||is mounted on a heavy -duty tubular stainless steel stand with floor flanges for level adjustment.|
|1. Door interlock||door safety locking device keep door from being opened while chamber is a pressurized during cycle.|
|2. Safety valve:||releasing steam in case of excessive chamber pressure|
|3. Low water protection device||low water level in steam generator will actuate audible alarm and protective device cut off the heating source immediately.|
|4. Circuit protection device:||circuit breaker and fuse protects control units when short circuit or overload occurs.|
|5.Emergency exhuast||manual exhaust to went the load in the event of a power outage.|
|Option|| □Automatic feed water device|
□Digital temperature controller
Jacket, Chamber Door
& shell Material
|Stainless Stell SUS316||Stainless Stell SUS304|
The purpose of Slider Kit is to gather common slideshow-like jQuery functionalities (such as news sliders, photos galleries/sliders, carousels, tabs menus)
When tracking mouse movement, you usually need to know the actual position of the mouse pointer. The event object that is passed to the handler contains some information about the mouse coordinates.