- Two channel independent isolated output
- Auto-programs and manual adjustment functions
- Medium and low frequency whole field treatment
．Push the lid and replace by 9V battery. Don't make mistake on position and negative pole.
． Be sure one side of the cable is inserted in the output plug, and the other side is inserted in the electrodes.
．Choice to cure which location and use water or alcohol to clean it, then paste the electrodes on the location.
．To select a function: Adjust the function slid switch indicator to MA, MT, MM, B, C, M.
．Slower turn the output current control knob for channel 1 or 2, the indicator will light up while the unit is in operation.
．Once you turn on the power, it will execute function which the light indicated function position.
．After finishing above steps, adjust the output current until you feel comfortable.
．It will stop work and the light will turn off automatically ( except the power indicated light ) When time is up to indicated timer.
．Switch the output current intensity control knob to the "off" position to end your treatment.